Thrombosis of the innominate artery during prolonged cardiac support by right transaxillary Impella 5.0 device
We describe an unexpected complication in a 46-years-old man affected by ischemic cardiomyopathy who received a prolonged mechanical cardiac support by means of right transaxillary Impella 5.0. The postoperative course was complicated by recurrent septicemia from multi-durg resistant Acinetobacter baumanii. He underwent implantation of venoarterial extracorporeal membrane oxygenation because of refractory cardiac arrest, followed by Impella removal that has been complicated by embolic stroke. The likely source of the embolus was a large thrombus inside the innominate artery. After a long-standing right transaxillary device implantation, particularly in patients with intercurrent septic episodes, a CT angiographic scan should be planned to exclude arterial thrombosis before removal of the transarterial device.
Chung JS, Emerson D, Ramzy D, et al. A new paradigm in mechanical circulatory support: 100 patient experience. Ann Thorac Surg 2020;109:1370-7. DOI: https://doi.org/10.1016/j.athoracsur.2019.08.041
Boll G, Fischer A, Kapur NK, Salehi P. Right Axillary Artery Conduit Is a Safe and Reliable Access for Implantation of Impella 5.0 Microaxial Pump. Ann Vasc Surg 2019;54:54-9. DOI: https://doi.org/10.1016/j.avsg.2018.10.004
Bertoglio L, Katsarou M, Scandroglio M, et al. Surgical transaxillary placement of the Impella 5.0 ventricular assist device. J Card Surg 2019;34:92-8. DOI: https://doi.org/10.1111/jocs.13978
Nakamura K, Krishnan S, Mahr C, McCabe JM. First-in-Man Percutaneous Transaxillary Artery Placement and Removal of the Impella 5.0 Mechanical Circulatory Support Device. J Invasive Cardiol 2017;29:E53-9.
Jones TL, Kearney KE, McCabe JM. Prevalence and Predictors of Vascular Thrombus Formation After Percutaneous Axillary Artery Impella Insertion. Circ Cardiovasc Interv 2019;12:e008046. DOI: https://doi.org/10.1161/CIRCINTERVENTIONS.119.008046
Castillo-Sang MA, Prasad SM, Singh J, et al. Thirty-five day Impella 5.0 support via right axillary side graft cannulation for acute cardiogenic shock. Innovations (Phila) 2013;8:307-9. DOI: https://doi.org/10.1097/imi.0000000000000009
Ranganath NK, Moazami N, Staniloae CS, Hisamoto K. Successful embolic protection during temporary circulatory support device removal in a patient who required holding of anticoagulation for postoperative hemothorax. J Thorac Cardiovasc Surg 2018;156:e203-4. DOI: https://doi.org/10.1016/j.jtcvs.2018.05.041
- Abstract views: 203
- PDF: 116
Copyright (c) 2021 The Authors
This work is licensed under a Creative Commons Attribution-NonCommercial 4.0 International License.